Everything was great, the manual, the response time, the simplicity of the order, and the
Price. The only thing that I could possible say on a negative note would be that the manual I ordered was more for a service tech. There were a lot of schematic diagrams that didn't help me solve the problem. However I would order again and recommend the web sight to others.
very complete. acceptable resolution. details are a little unclear. is a manual note 8.
Text excerpt from page 12 (click to view)
No. Adjusting point 1 PURITY ADJ. Adjusting procedure/conditions 1. Receive the GREEN-ONLY signal. Adjust the beam current to about 500 ÂµA. 2. Maintain the purity magnet at the zero magnetic field and keep the static convergence roughly adjusted. 3. Observe the points a, b as shown in Fig. 4-1 through the microscope. Adjust the landing to the A rank requirements. 4. Orient the raster rotation to 0 eastward. 5. Tighten up the deflection coil screws. Â» Tightening torque: 108 Â± 20 N (11 Â± 2 kgf) 6. Make sure the CRT corners landing meet the A rank requirements. If not, stick the magnet sheet to correct it. Note: This adjustment must be done after warming up the unit for 30 minutes or longer with a beam current over 500 ÂµA. Waveform and others